Fineness of Hydraulic Cement by Air Permeability Apparatus. ASTM C 11 Standard Test Methods for Fineness of Hydraulic Cement by Air Permeability Apparatus – Download as PDF File .pdf), Text File .txt) or read. ASTM C Standard Test Methods for Fineness of Hydraulic Cement by Air- Permeability Apparatus. standard by ASTM International.
|Published (Last):||2 June 2010|
|PDF File Size:||11.83 Mb|
|ePub File Size:||4.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
Historical Version s – view previous versions of standard. Work Item s – proposed revisions of this standard. This test method astk determination of the fineness of hydraulic cement, using the Blaine air-permeability apparatus, in terms of the specific surface expressed as total surface area in square centimetres per gram, or square metres per kilogram, of cement. Two test methods are given: The Blaine air-permeability apparatus consists essentially of a means of drawing a definite quantity of air through a prepared bed of cement of definite porosity.
The permeability cell shall consist of a rigid cylinder, constructed of austenitic stainless steel. The disk shall be constructed of noncorroding metal, and shall fit the inside of the cell snugly. The plunger shall be constructed of austenitic stainless steel and shall fit into the cell.
The filter paper disks shall be circular, with smooth edges, and shall have the same diameter as the inside of the cell. The U-tube manometer shall be constructed according to the design indicated.
ASTM C Fineness of Hydraulic Cement | CTLGroup
The manometer shall be filled to the midpoint line with a nonvolatile, nonhygroscopic liquid of low viscosity and density. The timer shall have a positive starting and stopping mechanism. The calibration of the air permeability apparatus shall be made using the standard reference material.
The automated test method shall employ apparatus designed either aztm the principles of the Blaine air-permeability method or apparatus based on the air-permeability principles of the Lea and Nurse method. When the specific surface values determined by an automated apparatus are to be used for acceptance or rejection of cement, the method used shall comply with the qualification requirements.
When standardization is required in order c2004 achieve agreement between test method A and test method B, the apparatus shall be standardized according to the requirements prescribed. This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application.
ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
ASTM C204 – 17
Test Method A is the Reference Test Method using the manually operated standard Blaine apparatus, while Test Method B permits the use of automated apparatus that has in accordance with the qualification requirements of this test method demonstrated acceptable performance.
Although the test method may be, and has been, used for the determination of the measures of fineness of various other materials, it should be understood that, in general, relative rather than absolute fineness values are obtained.
However, the user should exercise judgement in determining its suitability with regard to fineness measurements of cements with densities, or porosities that differ from those assigned to Standard Reference Material No.
Mercury, or its vapor, has been demonstrated to be hazardous to health and corrosive to materials. Astk should be taken when handling mercury and mercury containing products. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard. Historical Version s – view previous versions of standard Work Item s – proposed revisions of this asfm Translated Version s: Link to Active This link will always route to the current Active version of the standard.